In this space, visitors are invited to post any comments, questions, or skeptical observations about Philo T. Farnsworth's contributions to the field of Nuclear Fusion research.
Subject: Re: Neutron Counter interface
Date: Oct 28, 10:47 pm
Poster: Richard Hull
On Oct 28, 10:47 pm, Richard Hull wrote:
>Unfortunately there are no discriminator adjustments available. But I did turn up the gain and the high voltage a tiny bit to get about 1 count per 3 minutes. I don't know whether it's right or wrong to do this.
A tiny bit is OK, but the bottom line is the calibration must remain intact.
It is important that the mrem remain intact! Remember that for the rem dose rate meters there are classically 8 neuts/sqcm/sec at 1 mrem for 2.45mev. This is your key to you data! Counts on the device mean nothing per se.
- Re: Neutron Counter interface - Carl Willis Oct 29, 1:23 am
- Re: Neutron Counter interface - Richard Hull Oct 29, 7:58 pm