In this space, visitors are invited to post any comments, questions, or skeptical observations about Philo T. Farnsworth's contributions to the field of Nuclear Fusion research.
Subject: Re: Thermocouple Meters/Data Mining
Date: Dec 20, 9:19 pm
Poster: Tom Ligon
On Dec 20, 9:19 pm, Tom Ligon wrote:
I'm not overly impressed with the 531 either. I use it because we inherited a bunch of equipment that uses it. Recently we have made a push to move to Convectron or ConvecTorr gages, primarily to extend either end of the useful pressure range. We have one Piranni gage which is also better than the 531.
Generally the 531 gives out at about the point where vacuum starts getting interesting. The one on my Fusor drifted about 25 millitorr over a year of use, and it can be deucedly hard to get access to a known hard vacuum source to rezero a gage which is drifting.
- Re: Thermocouple Meters/Data Mining - Richard Hester Dec 21, 9:06 pm